+0,1 °C / - 0,1 °C in the range of 34°C bis 43 °C; außerhalb dieses Bereiches +0,2 °C / - 0,2 °C
Medical thermometer for determining body temperature
rectal, oral, axillary
-25 °C to 60 °C
Please refer to the instructions for use for full technical data.
Questions and answers
No, the Geratherm® flex cannot be switched between Celsius and Fahrenheit. The unit of measurement used is ° Celsius.
Yes, the battery can be changed.
Caution: Keep the thermometer and batteries away from small children. If the battery is swallowed, contact a hospital immediately. Do not throw the battery into fire, it may explode.
You can put a hygienic disposable cover on the probe to avoid virus transmission. When using disposable cases, there may be a deviation of 0.1 °C (0.2 °F) from the actual temperature. Discard the disposable sleeves after the measurement.
Disposable covers are available from the pharmacy.
Wipe the probe head with a solution of mild soapy water and cold water. Disinfect the thermometer by wiping the sensor and the lower section of the housing with a cloth previously dipped in alcohol (recommendation: isopropyl alcohol 70%). Do not boil the thermometer and do not clean it in the autoclave.
The Geratherm® flex is not waterproof.
The Geratherm® flex complies with IP 22 and is therefore protected against solid foreign bodies with a diameter of 12.5 mm and more and against water dripping at an angle of up to 15 degrees.