The Geratherm® easy med is a blood pressure monitor for the upper arm with a direct classification of the measured value according to the World Health Organization (WHO) rating system using a traffic light scale on the right side of the display. The blood pressure monitor can be operated with batteries or a mains adapter.
The device uses a traffic light scale to classify your measured value directly according to the WHO rating system.
In addition to batteries, Geratherm® easy med can also be operated with the supplied power supply.
You can save 3 x 30 readings in the device.
|Measuring range||0 mmHg - 299 mmHg|
|Measurement accuracy||+- 3 mmHg (0,4 kPa) from 5°C to 40°C|
|Measurement method||Oscillometric method|
|Intended purpose||Determination of systolic and diastolic blood pressure and heart rate|
|Power supply||4x Battery AA (1,5V DC)|
|Cuff size||23 - 33 cm|
|Weigh||approx. 265 g inkl. Batteries|
|Dimensions||approx. 110 x 150 x 80 mm|
|Please refer to the instructions for use for full technical data.|
Arm circumference: 23 cm to 33 cm
Arm circumference: 32 cm to 42 cm
Yes, the Geratherm® easymed displays this. But be careful, the display indicates that a pulse irregularity as a result of an irregular heartbeat was detected during the measurement. This is usually not a cause for concern.
However, if this symbol appears frequently, we recommend that you seek medical advice.
The display indicates that a pulse irregularity as a result of an irregular heartbeat was detected during the measurement. This is usually NOT a cause for concern. However, if this symbol appears frequently, we recommend that you seek medical advice. Please note that the device does not replace a cardiological examination, but is used to detect pulse irregularities at an early stage.
Yes, the batteries can be changed. 4x (AA) 1.5V alkaline batteries
Only use the power pack offered by Geratherm which is compatible with the device. You can purchase this from your specialist dealer.
ATTENTION! The product may only be operated with a medically approved, stabilized AC power pack (input: 230 V, AC, 50 Hz; output: 6 V, DC, 1 A).
1. No batteries are required when operating with the power supply unit.
2. If the current flow from the AC adapter is interrupted during measurement, the sphygmomanometer must be reset by disconnecting the AC adapter from the sphygmomanometer.
3. Only use medically approved power supplies that meet the specifications in this manual. Using other power packs may damage your blood pressure monitor.
The measurement can be taken on the left as well as on the right arm. However, further measurements should always be taken on the same arm. Preferably on the arm where the higher blood pressure is to be measured.
Individual blood pressure changes several times a day. It is influenced by a wide variety of factors. Therefore, please make the measurements under the same conditions.
Please do not use this device if you are allergic to polyester, nylon or plastic.
Before using the device by pregnant women, consult a doctor and assess the suitability of the measuring method. In some cases there may be deviations in measurement. Pregnant women are therefore recommended to have regular check measurements carried out by their doctor, since the clinical validation was not carried out with pregnant women.
To clean the case, use a cloth dampened with water or neutral detergent, and then wipe dry. Avoid thinners, petrol and other aggressive cleaning agents.